The Elcometer range of digital coating thickness gauges has been specifically designed to provide highly accurate, reliable and repeatable coating thickness. The Elcometer Coating Thickness. Gauge is an incredibly versatile gauge. With a range of probes in both Integral or Separate probe versions for coating. Elcometer Coating Thickness Gauge. Operating Instructions. English. R. English Equipment described in these instructions is covered by the following.
|Published (Last):||6 November 2006|
|PDF File Size:||8.62 Mb|
|ePub File Size:||8.86 Mb|
|Price:||Free* [*Free Regsitration Required]|
Battery life is reduced when using the back light. Postage cost can’t be calculated. The display will show PrF.
Elcometer Inc, Michigan Office Relocation. This item may be a floor model or an item that has been returned to the seller after a period of use. The backlight is designed for low light operations and can be turned on and off via the keypad. Consult your local Elcometer supplier. Gauges will have reduced range when adjusted for thin substrates. See all condition definitions – opens in a new window or tab Press to accept the calibration or wait 7 seconds for the gauge to accept the calibration automatically.
If is pressed the gauge displays the previous statistical value on the user list for a minimum of 3 seconds.
Elcometer NB Coating Thickness Gauge | eBay
In this routine if, after the key-lock feature is operated, 7 elcimeter elapses without pressing a key or taking a reading, the gauge will automatically continue to the next step, i.
This may be due to incorrect substrate, incorrect values or units, etc. Clear the error message and repeat the adjustment procedure. Two calibration foils are required; one below elco,eter thickness to be measured the thin foil and one above the thick foil. This listing was ended by the seller because there was an error in the listing. Then take a reading on the same bare substrate and adjust the gauge to read 10 m 0.
Coating and Thickness
Order using sales part elcometdr TEST Press and hold for 4 seconds to set the display to zero. To Recall The Factory Calibration. Press to exit menu or exit automatically when all 5 items have been entered.
The feature is switched on by selecting at least one elconeter value to be wlcometer – see Setting The User List For Statistical Values on page The bleeper will sound 3 times to indicate that this routine is complete. Learn More – opens in a new window or tab Any international postage and import charges are paid in part to Pitney Bowes Inc.
The gauge does not contain any user-serviceable components. The menu from which the list is created is as follows: If the statistics feature is switched on, the display will show one of the statistical values Statistical summary on page The display will blink and then re-display the reading.
Find a reebok cross trainer in United Kingdom on Gumtree, To delete the last reading from the statistical calculations when the statistics feature is switched on, press and hold for elcometrr than 2 seconds but less than 4 seconds. Seller assumes all responsibility for this listing. An intermediate foil thickness can also be used to confirm the performance over the calibrated range.
Your gauge can display statistical summary information in a preset sequence. To maximise the benefits of your new Elcometer Coating Thickness Gauge elcmoeter take some time to read these Operating Instructions. Contact Elcometer or your local Elcometer supplier. Switch off the gauge.